The following configuration instructions apply to the new emonPi_CM firmware only.
This firmware is available in two variants:
Recommended: LowPowerLabs radio format.
JeeLib Native radio format
The firmware does not support receiving data from the original JeeLib Classic radio format.
Make sure that all sensor nodes (e.g emonTx, emonTH) that send data to the emonPi_CM are running firmware that transmits in the correct format.
The emonPi_CM front-end firmware - description
The firmware that runs in the Atmel ATMega 328P primarily provides the energy monitoring function and handles the radio traffic coming in from other sensor nodes.
It uses emonLibCM, set up for two current channels, to obtain and process current and voltage readings and, from external sensors, temperature readings and the pulse input.
The radio traffic is almost completely processed within the RFM69CW radio module, all the
sketch needs to do is retrieve the complete message when it becomes available. When a
complete message is available, a call to
radio.receiveDone() will return
true, and the
message is formatted and sent via the serial connection to the Raspberry Pi.
Likewise, when a set of readings is available from emonLibCM,
returns true and the power values (or assumed apparent powers calculated from the
nominal voltage and measured current) are loaded into the emonpi structure and similarly
passed to the Raspberry Pi by
It is now possible to fully calibrate the emonPi. Calibration commands come in via the
same serial interface from the Raspberry Pi, and are handled by
getCalibration() (in a
separate source file).
It’s also possible to transmit a message – that function is handled by the outbound r.f. data handler.
Configuration, calibration and the latest energy values are stored in the EEPROM memory of the ATMega 328P. The task of saving the data to, and retrieving the data from, the EEPROM is handled
Configuration & on-line calibration
On-line configuration is locked for the first 3 minutes after the front-end sketch has started in order to allow the Raspberry Pi to start, without any data emitted to the serial port affecting the configuration or calibration.
If the emonPi does not appear in emonCMS the very first time it is powered up (following setting up the Wi-Fi parameters etc), restart the energy monitoring front end with a short press of the reset button adjacent to the CT1 connector.
The following setup/configuration commands are available:
l list the settings (terse)
L list the settings (verbose)
r restore sketch defaults & restart
s save config to EEPROM
v show firmware version
V verbose mode - changes etc echoed to serial output
b<n> set r.f. band n = a single numeral: 4 = 433MHz, 8 = 868MHz, 9 = 915MHz (may require hardware change)
p<nn> set the r.f. power. nn - an integer 0 - 31 representing -18 dBm to +13 dBm. Default: 25 (+7 dBm)
g<nnn> set Network Group nnn - an integer (OEM default = 210)
n<nn> set node ID n= an integer (standard node ids are 1..60)
c<n> n = 0 for OFF, n = 1 for ON, enable current & power factor values to serial output for calibration.
d<xx.x> a floating point number for the datalogging period
k<x> <yy.y> <zz.z>
Calibrate an analogue input channel:
x = a single numeral: 0 = voltage calibration, 1 = ct1 calibration, 2 = ct2 calibration, etc
yy.y = a floating point number for the voltage/current calibration constant
zz.z = a floating point number for the phase calibration for this c.t. (z is not needed, or ignored if supplied, when x = 0)
e.g. k0 256.8
k1 90.9 2.00
f<xx> the line frequency in Hz: normally either 50 or 60
a<xx.x> a floating point number for the assumed voltage if no a.c. is detected
m<x> <yy> meter pulse counting:
x = 0 for OFF, x = 1 for ON
yy = an integer for the pulse minimum period in ms. (y is not needed, or ignored when x = 0)
t0 <y> turn temperature measurement on or off: y = 0 for OFF, y = 1 for ON, y = 2 ON & search for sensors
t<x> <yy> <yy> <yy> <yy> <yy> <yy> <yy> <yy>
change a temperature sensor’s address or position:
x = a single numeral: the position of the sensor in the list (1-based)
yy = 8 hexadecimal bytes representing the sensor’s address
e.g. 28 81 43 31 07 00 00 D9
N.B. Sensors CANNOT be added beyond the array size.
T<ccc>\n - transmit a string of alpha-numeric characters.
w<x> set the radio on or off 0:OFF, 1:Receive ON, 2:Transmit ON, 3:both ON
z set the energy values and pulse count(s) to zero
? show helper text
The command will be acknowledged, and confirmation given when saving the settings, only after verbose mode has been set. When you change one or more of the settings, the change will take effect immediately and you might see the displayed values change. (See below for the temperature sensors, which are different.)
Option (‘s’) will save all the changes. If you do not do this, the settings will revert to the previous values at the next restart. After you save (‘s’) the changes, the new settings will be used forever, or until changed again.
If you restore the sketch default values (‘r’), all the EEPROM data is ignored and the sketch restarts immediately, using the values set in the sketch. There is then no means of recovering the EEPROM data.
If the EEPROM has been used previously and has had non-compliant values written, the EEPROM content will be ignored and the sketch will start using its own default values. Saving the configuration will format the EEPROM and the sketch’s set values will be stored.
Adding or moving Temperature Sensors
Sensors can be added, or moved around, without stopping continuous monitoring of power and energy, but temperature readings must be disabled. If you are indifferent as to the order of the sensors, then all you need to do is stop temperature monitoring with ‘t0 0‘ followed by ‘t0 1’ to restart it. The sensors will be detected and read. If you need to preserve the order which they appear, save the configuration to EEPROM with ‘s’.
If you need to preserve the order of existing sensors, or put new sensors into a specific order, then the procedure is:
Make a note of the addresses (serial numbers) and the order of any existing sensors with the verbose ‘V’ then list ‘L’ commands.
Stop temperature monitoring with ‘t0 0’
Connect the new/replacement sensor(s) and enable temperature monitoring with ‘t0 1’, list the sensors again.
You will probably find the order of sensors is not what you want. You can put the sensors into the order you need by typing in the address of each. Stop temperature monitoring with ‘t0 0’, then type each sensor address into the position it needs to occupy, e.g:
‘t1 28 81 43 31 07 00 00 D9’
‘t2 28 8D A5 C7 05 00 00 D5’
(‘28’ in the first byte is the signature of a DS18B20.)
Enable temperature monitoring with ‘t0 1’, list the sensors and check that all is correct and that they are reading as expected.
Save the configuration to EEPROM with ‘s’